Huber needles used in implanted ports to withdraw blood, inject medications, and other solutions.
Washington, DC: The U.S. Food and Drug Administration today announced a Class I recall of Exel/Exelint Huber needles, Exel/Exelint Huber Infusion Sets and Exel/Exelint “Securetouch+” Safety Huber Infusion Sets, manufactured by Nipro Medical Corporation for Exelint International Corporation.
Huber needles are used to access ports implanted under the skin of chronically ill patients for repeated access to veins for the withdrawal of blood and infusion of medication, nutritional solutions, blood products, and imaging solutions. These needles should be designed to penetrate the port without cutting and dislodging any silicone cores (or slivers) from the ports into which they are inserted.
Inspections conducted in October 2009 of Nipro facilities in Japan found that their needles “cored” in 60 to 72 percent of tests. The reason for this coring is related to design and manufacturing processes, which the FDA continues to investigate.
There are more than 2 million units impacted by this recall in distribution nationwide. Recalled needles were manufactured from January 2007 to August 2009. Units subject to recall have a lot number that begins with “07,” “08,” “09,” ; for a complete list of product codes or catalog numbers see the link below.
Following hospital reports to the FDA of leakage after accessing the port with a Huber needle (labeled to be non-coring), the agency conducted it own laboratory testing of Huber needles from multiple manufacturers. This testing showed that certain Huber needles produced cores when inserted into ports. However, at this time only needles manufactured by Nipro have shown a high frequency for coring. The agency is continuing its investigation and will update the public if there are new developments.
At this time, the FDA has not received any adverse event reports related to silicone foreign bodies released in patients from Huber needle coring.The agency received only reports of port leakage. However, because it may be very difficult for clinicians to associate adverse patient outcomes with the use of defective Huber needles, there may be under-reporting of events.
JAN-26-10: FDA Announces Class I Recall of Certain Infusion Set Needles [FDA: RECALLS]
If you or a loved one has suffered an adverse health event resulting from the use of this product; click on title above for legal help and a free evaluation of your possible case;
EXPOSING the FDA and the USDA - Broad Casting here the things that they would prefer us NOT to know about our FOOD & DRUGS & Farming.